The process to achieve this is loop through this code, but changing the voltage range and hence reading back different data. Here with missing subvi,as u told its nit at all starting with initial voltage and ending at final voltage, but i m getting reading for -1 v and 1v using my labview program. Message 22 of Your mentioned vi is for only increment it seems, But I need within the voltage range -1v to 1v we need to measure current across the 1 kiloohm resistor, which we have connected as in Keithley Manual. But problem is each time I’ve to mention. Message 29 of
|Date Added:||21 May 2015|
|File Size:||21.99 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Its in labview 5.
Most Active Hardware Boards: You can set breakpoints and single step through the code to see what the instrument is actually being programmed for and whether it is triggering or not. I m sending what I m using here, so if u try for increment in my code.
Message 3 of Sir you just go through labvieew attachment in that you see V-Source I’ve mentioned Initial Voltage and Final Voltage middle increment window and if you have any correction then do it and send it to me sir.
I do not have the instrument and you did not post a critical subVI.
Keithley 6517B User Manual: Overview; Software Support; Labview Driver; Remote Interfaces
They are not decorations. Make change and forward it to me.
You don’t get it. U just simplify my vi which i sent before, Only problem is i m not at all getting readings between -1 to 1v.
You can ,eithley manually operate the instrument and try to take a measurement. I need to work on this. After a long gap, i m writing to you, coz i tried alll the ways to solve my problem,but all ways not at all working, so now i nee dyour help.
Sorry sir, this is only subvi. You posted the same VI without the subVI. For hi-jacking thread, I was searching related to my problems, I found that one, labviee there only I asked my question. Please show your latest VI and typical results for the critical range of Message 6 of But i need such code’s through that i m able to apply for my 5.
How to add decimal point in out put data – Page 3 – Discussion Forums – National Instruments
Message 9 of We are getting reading from meter. For that we have created code. Keithley xy graph.
This code will generate the correct values that you would input to the instrument. Posting a top level VI does not mean that you attach all of the kdithley.
You should post the code you aer using now. Never getting the desired result.
Labview VI’s for keithley meter – Discussion Forums – National Instruments
Check keothley kind of answer you get from the device as asked before: So may be it will difficult you to open. Kfithley the things are ok. Nobody can help you on such vague information Message 10 of I m going to attach my new vi. Are you using the VIs which I found here, at ni. Your mentioned vi is for only increment it seems, But I need within the voltage range -1v to 1v we need to measure current across the 1 kiloohm resistor, which we have connected as in Keithley Manual.